Osciloscop Digital DSO4202C Hantek
DSO4000C Series :
2 Channel Digital Oscilloscope + Arbitrary/Function Waveform Generator + Synchronizing Signal + External Trigger.
The Keys for osilloscope and waveform generator is seperated for convenient to operate it simultaneously.
Large and clear display (7 inch 64K color LCD display, Resolution 800x480), more clear and realistic waveform.
Oscilloscope: 1GSa/s Sample Rate, 200/100/70MHz Bandwidth, more than 20 kinds of auto measurement functions.
Powerful Trigger Function: Edge, Video, Pulse, Slope, Over time, Alternative.
Arbitrary/Function Waveform Generator: 25MHz, 12 bits resolution, 200MHz DDS, Arbitrary wave/square wave/sine wave/triangle wave/trapezoidal wave/pulse wave/DC is easy to simulate transducer.
Support SD card, VGA function (optional).
Integrated USB Host, Support USB disk storage, USB interface/SD card system update.Features :
100MHz Bandwidth; 1GSa/s Sample Rate.
2 Channel Oscilloscope; 40K Record Length.
7 inch 64K color LCD display, Resolution 800x480.
32 kinds of Automotive measurement, with FFT function.
25MHz Arbitrary waveform output (Sine wave up to 75MHz).
Powerful trigger function: Video, Edge, Pluse Width, Slope, Overtime, Alternate Trigger.
|Sample Rate||Sampling Rate Range: 1GSa/s|
|Equivalent Sample Rate: 25GSa/s|
|Normal||Normal data only|
|Peak Detect||High-frequency and randon glith capture|
|Average||Wavefom Average, selectable 4,8,16,32,64,128|
|Inputs Coupling||AC, DC, GND|
|Inpits Impendance||1M2% 20pF3pF|
|Probe Attenuation||1X, 10X|
|Supported Probe Attenuation Factor||1X, 10X, 100X, 1000X|
|Maximum Input Voltage||CAT I and CAT II: 300VRMS (10×)|
|CAT III: 150VRMS(1×);|
|Installation Category II: derate at 20dB/decade above 100kHz to 13V peak AC at 3MHz* and above. For non-sinusoidal waveforms, peak value must be less than 450V. Excursion above 300V should be of less than 100ms duration. RMS signal level including all DC components removed through AC coupling must be limited to 300V. If these values are exceeded, damage to the oscilloscope may occur.|
|Sample Rate Range||1GS/s|
|Waveform Interpolation||(sin x)/x|
|SEC/DIV Range||4ns/div to 40s/div|
|Sample Rate and||50ppm at over any 1ms time interval)|
|Delay Time Accuracy|
|Position Range||20ns/div to 80us/div; (-8div x s/div) to 40ms;|
|200us/div to 40s/div; (-8div x s/div) to 400s|
|Delta Time Measurement Accuracy
|Single-shot, Normal mode: (1 sample interval +100ppm × reading + 0.6ns);
>16 averages: (1 sample interval + 100ppm × reading + 0.4ns);
Sample interval = s/div ÷ 200
|Vertical Resolution||8-bit resolution, all channel sampled simultaneously|
|Position Range||2mV/div to 10V/div|
|Rise Time at BNC( typical)||3.5ns|
|Analog Bandwidth in Normal and Average modes at BNC or with probe, DC Coupled||500mV/div to 2V/div, 40V; 5V/div, 50V|
|Math||+, -, *, /, FFT|
|FFT||Windows: Hanning, Flatop, Rectamgular, Bartlett, Blackman;|
|1024 sample point|
|Low Frequency Response (-3db)||10Hz at BNC|
|DC Gain Accuracy||4% for Normal or Average acquisition mode, 5mV/div to 2mV/div|
|DC Measurement Accuracy,
Average Acquisition Mode
|When vertical displacement is zero, and N 16: (3% × reading + 0.1div + 1mV) only 10mV/div or greater is selected;
When vertical displacement is not zero, and N16: [3% × (reading + vertical position) + 1% of vertical position + 0.2div]; Add 2mV for settings from 2mV/div to 200mV/div; add 50mV for settings from 200mV/div to 5V/div
|Volts Measurement Repeatability, Average Acquisition Mode||Delta volts between any two averages of 16 waveforms acquired under same setup and ambient conditions|
|Trigger Types||Edge, Video, Pulse, Slope, Over time, Alternative|
|Trigger Source||CH1, CH2, EXT, EXT/5, AC Line|
|Coupling Type||DC, AC, Noise Reject, HF Reject, LF Reject|
(Edge Trigger Type)
|1div from DC to 10MHz; 1.5div from 10MHz to 100MHz; 2div from 100MHz to Full;|
|200mV from DC to 100MHz; 350mV from 100MHz to 200MHz;|
|1V from DC to 100MHz;1.75V from 100MHz to 200MHz;|
|AC: Attenuates signals below 10Hz;|
|HF Reject: Attenuates signals above 80kHz;|
|LF Reject: Same as the DC-coupled limits for frequencies above 150kHz; attenuates signals below 150kHz|
|Trigger Level Range||CH1/CH2: 8 divisions from center of screen;|
|Trigger Level Accuracy( typical)Accuracy is for signals having rise and fall times 20ns||CH1/CH2: 0.2div × volts/div within 4 divisions from center of screen;|
|EXT: (6% of setting + 40mV);|
|EXT/5: (6% of setting + 200mV);|
|Set Level to 50%(typical)||Operates with input signals 50Hz|
|Video Trigger Type||CH1, CH2: Peak-to-peak amplitude of 2 divisions;|
|Signal Formats and Field Rates, Video Trigger Type||Supports NTSC, PAL and SECAM broadcast systems for any field or any line|
|Holdoff Range||100ns ~ 10s|
|Pulse Width Trigger|
|Pulse Width Trigger Mode||Trigger when (< , >, = , or ); Positive pulse or Negative pulse|
|Pulse Width Trigger Point||Equal: The oscilloscope triggers when the trailing edge of the pulse crosses the trigger level.
Not Equal: If the pulse is narrower than the specified width, the trigger point is the trailing edge. Otherwise, the oscilloscope triggers when a pulse continues longer than the time specified as the Pulse Width.
Less than: The trigger point is the trailing edge.
Greater than (also called overtime trigger): The oscilloscope triggers when a pulse continues longer than the time specified as the Pulse Width
|Pulse Width Range||20ns ~ 10s|
|Slope Trigger Mode||Trigger when (< , > , = , or ); Positive slope or Negative slope|
|Slope Trigger Point||Equal: The oscilloscope triggers when the waveform slope is equal to the set slope.|
|Not Equal: The oscilloscope triggers when the waveform slope is not equal to the set slope.|
|Less than: The oscilloscope triggers when the waveform slope is less than the set slope.|
|Greater than: The oscilloscope triggers when the waveform slope is greater than the set slope.|
|Time Range||20ns ~ 10s|
|Over Time Mode||Rising edge or Falling edge|
|Time Range||20ns ~ 10s|
|Trigger on CH1||Internal Trigger: Edge, Pulse Width, Video, Slope|
|Trigger on CH2||Internal Trigger: Edge, Pulse Width, Video, Slope|
|Trigger Frequency Counter|
|Readout Resolution||6 digits|
|Accuracy (typical)||30ppm (including all frequency reference errors and 1 count errors)|
|Frequency Range||AC coupled, from 4Hz minimum to rated bandwidth|
|Signal Source||Pulse Width or Edge Trigger modes: all available trigger sources|
|The Frequency Counter measures trigger source at all times, including when the oscilloscope acquisition pauses due to changes in the run status, or acquisition of a single shot event has completed.|
|Pulse Width Trigger mode: The oscilloscope counts pulses of significant magnitude inside the 1s measurement window that qualify as triggerable events, such as narrow pulses in a PWM pulse train if set to < mode and the width is set to a relatively small time.|
|Edge Trigger mode: The oscilloscope counts all edges of sufficient magnitude and correct polarity.|
|Video Trigger mode: The Frequency Counter does not work.|
|Cursor Measurement||Voltage difference between cursors: V|
|Time difference between cursors: T|
|Reciprocal of T in Hertz (1/T)|
|Auto Measuerment||Frequency, Period, Mean, Pk-Pk, Cycli RMS, Minimum, Maximum, Rise time, Fall Time,|
|+Pulse Width, -Pulse Width, Delay1-2Rise, Delay1-2Fall, +Duty, -Duty, Vbase, Vtop, Vmid,|
|Vamp, Overshoot, Preshoot, Preiod Mean, Preiod RMS, FOVShoot, RPREShoot, BWIDTH,|
|FRF, FFR, LRR, LRF, LFR, LFF|
|Signal Source Mode|
|Output Waveform||Arbitrary wave/square wave/sine wave/triangle wave/trapezoidal wave/pulse wave/DC|
|Waveform Range||3.5V Max|
|Output Current||50mA Ipeak=100mA|
|Harmonic Distortion||-50dBc 1KHz -40dBc 10KHz|
|Display Type||7 inch 64K color TFT (diagonal liquid crystal)|
|Display Resolution||800 horizontal by 480 vertical pixels|
|Display Contrast||Adjustable (16 gears) with the progress bar|
|Probe Compensator Output|
|Output Voltage( typical)||About 5Vpp into 1M load|
|Supply Voltage||100-120VACRMS(10%), 45Hz to 440Hz, CAt II|
|120-240VACRMS(10%), 45Hz to 66Hz, CATII|
|Fuse||2A, T rating, 250V|
|Temperature||Operating: 32 to 122 (0 to 50);|
|Nonoperating: -40 to 159.8(-40 to +71)|
|Humidity||+104 or below (+40 or below): 90% relative humidity;|
|106to 122(+41 to 50): 60% relative humidity|
|Altitude||Operating: Below 3,000m (10,000 feet);|
|Nonoperaring: Below 15,000m(50,000 feet)|
|Size||Length 385mm, Width 200mm, Height 245mm|
|Weight||3.5KG(with Packing); 2.08KG(without Packing|
Package includeds :
1 x Hantek DSO4102C Handheld Digital Multimeter Oscilloscope
2 x Passive Probe
1 x Alligator Clip Cable
1 x USB Cable
1 x CD
1 x User Manual
Details pictures :